Provides age-adjustable fever guidance to take the guess work out of interpreting temperature for the whole family.
Patented pre-warmed tip
Designed to minimize the cooling effect of the tip inside the ear canal, providing accurate measurements time after time.
Large color-coded display provides instant fever detection for the whole family.
Night mode with light and silent mode
ThermoScan® 7+ features a light for easy use in the dark and silent mode for avoding wake-ups.
ExacTemp® stablity indicator
ExacTemp® technology uses a light and beep to confirm proper positioning.
Safe and hygienic
Braun disposeable lens filters help to eliminate the risk of cross-contamination and ensure accurate readings.
How it works
Braun ThermoScan® measures the infrared heat generated by the eardrum and surrounding tissue. Because the eardrum shares blood supply with the temperature control centre in the brain, measurement with Braun ThermoScan® accurately reflects changes in core body temperature.
Guaranteed accuracy for all ages
With its patented pre-warmed soft tip and guidance system, the Braun ThermoScan® ear thermometer is suitable for babies ears from birth onwards.
No lens filter is attached.
Attach new, clean lens filter.
The thermometer probe was not positioned securely in the ear. An accurate measurement
POS = position error
Take care that the positioning of the probe is correct and remains stable.
Replace lens filter and reposition. Press Start button to begin a new measurement.
Ambient temperature is not within the allowed operating range
Allow the thermometer to remain for 30 minutes in a room where
Temperature taken is not within typical human temperature range (34 – 42.2 °C or 93.2 – 108 °F).
HI = too high
Make sure the probe tip and lens are clean and a new, clean lens filter is attached. Make sure the thermometer is properly inserted. Then, take a new temperature.
System error – self-check display flashes continuously and will not be followed by the ready beep and the ready symbol.
If error persists,
If error still persists,
Wait 1 minute until the thermometer turns off automatically, then turn
... reset the thermometer by re-moving the batteries and putting them back in.
... please contact the service centre.
Battery is low, but thermometer will still operate correctly.
Insert new batteries.
Battery is too low to take correct temperature measurement.
Insert new batteries.
Care and cleaning
The probe tip is the most delicate part of the thermometer.
It must be clean and intact to ensure accurate readings.
If the thermometer is ever accidentally used without a lens filter, clean the probe tip as follows: Very gently wipe the surface with a cotton swab or soft cloth moistened with alcohol. After the alcohol has completely dried out, you can put a new lens filter on and take a temperature measurement.
If the probe tip is damaged, please contact the local service center.
Use a soft, dry cloth to clean the thermometer display and exterior. Do not use abrasive cleaners. Never submerge this thermometer in water or any other liquid. Store thermometer and lens filters in a dry location free from dust and contamination and away from direct sunlight.
Additional lens filters (LF 40) are available at most stores carrying Braun ThermoScan.
Replacing the batteries
The thermometer is supplied with two 1.5 V type AA (LR 06) batteries. For best performance, we recommend Duracell® alkaline batteries.
❶ Insert new batteries when the battery symbol appears on the display.
❷ Open the battery compartment. Remove the batteries and replace with new batteries, making sure the poles are in the right direction.
❸ Snap battery door into place.
ExacTemp® stability indicator
Memory storage capacity
Easy-to-read LCD display
Protective case with lens filter storage
1.5V AA LR6 2x
Disposable lens filter 21x